GAC® 2100 Agri Product Support
Below are the support materials available for the GAC® 2100 Agri.
A: The GAC2100 Agri can not.
A: The Agri holds 8 grains.
A: The GAC2100 Agri takes around 33 seconds to test a sample.
A: The operating temperatures of the machines are 41-113˚.
A: The tester can test grains with a range from 32- 104˚.
A: No, both testers will display moisture, temperature and test weight.
A: On startup when asked which language choose number 8 English.
A: An error 9 means the sample temperature exceeds its limit. Allow the sample to cool down or warm up, and then try again. The machine and the grain temperature need to be within 20˚ of each other or you receive an error 9 message.
A: You need a printer that meets the following requirements: Must be a strip printer with an RS-232 serial port connection 1200 baud rate, 8 data bits, no parity and 1 stop bit.
A: No, from the main menu choose 3 Calibration Menu, then 2. Enter/ change Calibration, select the grain to change. Depress the white button on the rear lower right corner (Note NTEP machine are sealed and can only be changed by a state certified company or personnel). Then select the K numbers that need to be changed.
- Error Codes
- Spanish Operator's Manual
- DICKEY-john Moisture Testers Brochure
- Hot and Cold Corn Calibrations 12-15
- July 2017 Moisture Meter Calibration Constants
- 201706 GAC 2100 Blue Gray AGRI Operators Manual
- GAC 2100 Manual 09_01
- GAC2100 Spec Sheet A4
- GAC2100 Spec Sheet DE
- GAC2100 Spec Sheet FR
- GAC2100 Spec Sheet US
You must have Adobe Reader in order to access PDF files. Download Adobe Reader.